Titanium surface modification by using microwave-induced argon plasma in various conditions to enhance osteoblast biocompatibility
- Gyeung Mi Seon†1, 2,
- Hyok Jin Seo†1, 2,
- Soon Young Kwon1,
- Mi Hee Lee1,
- Byeong-Ju Kwon1, 2,
- Min Sung Kim1, 2,
- Min-Ah Koo1, 2,
- Bong Joo Park3 and
- Jong-Chul Park1, 2Email author
© Seon et al.; licensee BioMed Central. 2015
Received: 1 December 2014
Accepted: 11 April 2015
Published: 10 May 2015
Titanium is a well proven implantable material especially for osseointegratable implants by its biocompatibility and anti-corrosive surface properties. Surface characteristics of the implant play an important role for the evolution of bone tissue of the recipient site. Among the various surface modification methods, plasma treatment is one of the promising methods for enhance biocompatibility. We made microwave-induced argon plasma at atmospheric pressure to improve in titanium surface biocompatibility.
Various states of emission spectra from excited species-argon, nitrogen atoms and oxygen atoms were observed. The electron energy band structures are the unique characteristics of atoms and functional groups. Microwave-induced argon plasma treatment changed the titanium surface to be very hydrophilic especially on the 5 s short treatment and 30 s, 90 s long treatment samples that detected by contact angle measurement. MC3T3-E1 attachment and proliferation assay significantly increased in 5 s at short treatment, 30 s, and 90 s at long treatment after 5 days incubation.
Result indicated that microwave-induce argon plasma treatment would be an effective method to modify titanium surface for enhancing cell-material interactions.
KeywordsTitanium Microwave-induced argon plasma Surface modification Biocompatibility
The success of a dental implant is based on the osseointegration that is defined as the direct contact between the bone tissue and the dental implant surface, without fibrous tissue growing at the interface [1,2]. Titanium is a well proven implantable material especially for dental and orthopedic implants [3–5]. The popularity of titanium for osseointegratable implants is based on its biocompatibility and anti-corrosive surface properties. Surface characteristics of the implant play an important role for the evolution of bone tissue of the recipient site, after implantation [6,7]. In order to further enhance its biocompatibility, several special treatments to create surfaces such as sandblasted large-grit and acid-etched (SLA) surface, resorbable blasted media (RBM) surface and micro-arc oxidized (MAO) surfaces were conventionally used.
Among the various surface modification methods, plasma treatment is one of the promising methods. Diverse functional groups could be introduced to the surface by various appropriate plasma sources. These functional groups improve biocompatibility of the surface and enhance cellular responses . An atmospheric plasma system is commonly used for sterilization and surface modification. As previously, our laboratory made microwave-induced argon plasma at atmospheric pressure. This system provided good sterilization effects on microorganisms, fungi, Escherichia coli, Methicillin-resistant staphylococcus aureus and mycotoxins [9–12]. Also, surface modification effects on various biomaterials such as PLGA, polyurethane and silk fibroin were investigated by microwave-induced argon plasma [13–15].
In this study, microwave-induced argon plasma was adopted to enhance titanium surface biocompatibility, and different distance from nozzle and treating time were controlled to confirm the effects of this plasma system. Distances between sample and nozzle were set to 3 cm and 7 cm. For convenience, ‘short treatment’ stands for the test of 3 cm distance and ‘long treatment’ stands for the test of 7 cm distance.
Thickness and diameter of the CP-2 titanium (Ti) specimens (Seoul Titanium Inc., Korea) were 0.3 mm and 13 mm. The specimens were solvent-cleaned by a serial ultrasonic treatment in acetone, methyl alcohol and de-ionized water, and then dried with oxygen blow.
Treatment of microwave-induced argon plasma
Self-designed microwave-induced argon plasma was used in this study as previously described . Briefly, this system consisted of 2.45 GHz magnetron power supply (1 kW), an applicator including a tuning section and the nozzle made of quartz. The microwave was introduced through a WR-284 copper waveguide with internal cross section dimensions of 72 mm × 24 mm. The plasma generated at the end of a nozzle was formed by an interaction between the high electrical fields, which is generated by the microwave power supply, the waveguide aperture and the gas nozzle. The electric field intensity around the nozzle was calculated by a high frequency structure simulator (HFSS) code simulation as previously described . Argon was used as a working gas for this plasma system, which was chosen for its inertness, and the gas flow rate is approximately 100 min−1 at 8 kgf/cm2. For evaluating the plasma efficiency on Ti disks, Ti disks were placed in front of a nozzle in the plasma system and exposed to plasma for 1 s, 5 s at ‘short treatment (3 cm)’ and 10 s, 30 s, 90 s at ‘long treatment (7 cm)’, respectively.
Measurement of emission spectra
The emission spectra from the plasma according to the gas type were acquired through the optical fiber in the range of 200-1200 nm wavelength. The optical emission spectrometer (HR4000, Ocean Optics Inc, USA) was used to analyze the emission spectra and the electronically excited species generated by the plasma device. The measurement point has been set to be just region of plasma plume. The optical emission spectrum was obtained from the plasma device.
Contact angle measurement
Contact angle was measured using the sessile drop method . to evaluate hydrophilicity of prepared specimens. De-ionized water was utilized as probing liquid. Surface energy was also measured by same method. Three samples were used to collect the contact angle data in diiodomethane, formamide and glycerol for each plasma treated titanium. Lewis acids and bases were used to calculate the surface energy .
MC3T3-E1 mouse pre-osteoblastic cell line (Riken, Japan) was maintained in αMEM medium modified with ascorbic acid supplemented with 10 % fetal bovine serum and 1 % antibiotic antimycotic solution (WelGENE, Korea). Cells were cultured at 37 °C in a humidified 5 % CO2 atmosphere incubator.
Cell attachment and proliferation test
Microwave-induced argon plasma treated Ti disks were placed to 24 well plate and the titanium disks were sterilized in 70 % ethanol for 30 min and washed three times in distilled water. After sterilization, MC3T3-E1 cells were seeded on titanium disks at a density of 6 × 104 cells/disk and incubated for 2 h for cell attachment test. For cell proliferation test, cells were seeded at a density of 3 × 104 cells/disk and cultured for 1, 3, 5 days in a CO2 incubator. The viability of the cells was determined with 3-(4,5-dimethylthiazol-2-yl)-2,5-diphenyltetrazolium bromide (MTT) assay. Briefly, MTT was added to each well at 500 μg/mL and further incubated for 4 h at 37 °C. After incubation, dimethyl sulfoxide and glycine buffer was added to each well for the solubilization of the formed formazan salts. The absorbance of each well was measured spectrophotometrically at 570 nm by a microplate reader.
Cell morphometric analysis
For cell morphometric analysis, the cells were seeded at the initial seeding density of 1.0 × 104 cells/disk and after 2 h of incubation they were fixed with ice cold 70 % ethanol. Cell plasma membrane was visualized by staining with Texas Red C2-maleimide (Texas Red, 30 ng/ml in PBS, Invitrogen) . Cell nuclei were counterstained with Hoechst 33258 (1 g/ml in PBS, Sigma). Ten random pictures were taken by IX-70 microscope, equipped with a DP-71 digital camera (Olympus, Japan). Cell attachment area, perimeter and Feret’s diameter of 30 cells for each group were measured by Image J software (NIH, USA) .
All results were statistically analyzed by Student t-test. Data are expressed as mean ± standard error of mean. P values <0.05 were considered statistically significant.
Optical emission spectroscopy (OES)
Contact angle measurement
Long treatment condition; contact angle and surface energy on plasma treated titanium (n = 3)
Surface energy (mJ/m2)
Short treatment condition; contact angle and surface energy on plasma treated titanium (n = 3)
Surface energy (mJ/m2)
MC3T3-E1 attachment and proliferation
Cell morphometric analysis
The area, perimeter and Feret’s diameter of the attached MC3T3-E1 were measured and plotted in Figs. 5 and 6. All of the measured parameters increased significantly on the plasma treated titanium regardless of the treating time and treating condition.
Gaseous plasma provides a complex body of radicals, ions, electrons, and photons which are in various excited energy states. The optical emission spectra from the energy quenching processes of the excited species can be performed to characterize the plasma. The electron energy band structures are the unique characteristics of atoms and functional groups. Since the atmospheric pressure discharge was fired at ambient atmosphere, the emissions from not only excited argon atoms but also excited nitrogen and oxygen atoms and OH molecules could be detected. The emission spectra of the microwave-induced argon plasma were collected by optical emission spectroscopy (OES) in order to probe the existence of excited nitrogen derivatives. Compared to the short treatment, the long treatment could not be detected by OES. OES can observe the optical atmosphere and the long treatment cannot provide optical condition. Although emission spectra of the long treatment were not detected, radicals and electrons were still existed at the atmosphere. This phenomenon is easily understood by contact angle analysis.
Microwave-induced argon plasma treatment changed the titanium surface to be very hydrophilic. In general, surface hydrophilicity is very important for homogeneous and sufficient surface-compatibility as well as for good growth of cells on polymer materials . Furthermore, it was reported that enhanced fibronectin adsorption and reduced albumin adsorption occurs on hydrophilic surfaces and these elongated fibronectin conformation on the hydrophilic surface was effected to cell attachment . This plasma system could produce super-hydrophilic surface even for the short period time in the short treatment condition. In the long treatment condition, the surface had changed to super-hydrophilic after 90 s even though the OES was not detected at this point. In case of UV modification, 48 h treatment time is required to decrease the water contact angle of machined surface to less than 10 °. It might be resulted from the high population of various exited species that only short treatment time of microwave-induced argon plasma was required to achieve super-hydrophilic surface, which is a typical characteristic of atmospheric pressure gaseous discharge.
To confirm the effects of this plasma system for cellular responses, different treating conditions were adopted on this study. The initial interaction of cells with the biomaterial plays a key role for and early acceptance of the implant . In previous study, we confirmed that titanium surface treated helium atmospheric pressure glow discharge (He-APGD) promoted selectively higher absortion of fibronectin, a protein of critical importance for cell/biomaterial interaction. Therefore, we expected the improved cell response appeared to be dependent on the fibronectin-integrin mediated mechanism of cell attachment . In this sense, cell attachment and proliferation test was performed on the plasma treated titanium surface. Cell morphometric analysis is also performed to confirm changed cell morphology after plasma treatment. These results support that the microwave induced argon plasma system which was used for this study could be very useful to enhance cellular responses on titanium surface for both the short treatment and the long treatment. Even though this plasma has very strong power which can be used for sterilization microorganisms in a very short time, it can be controlled very easily by changing the position of samples.
Microwave-induced argon plasma system was employed to modify titanium surface in order to achieve enhanced biological responses. Plasma treated titanium surface turned to be very hydrophilic surface and it did not require more than 90 s treatment time. Surface modified titanium by plasma treatment increased surface energy and it allowed enhancing cell attachment and proliferation on the titanium surface. The increase of cell population, cell area, perimeter, and Feret’s diameter of MC3T3-E1 cells were also promoted. Conclusively, microwave-induce argon plasma treatment would be an effective method to modify titanium surface for enhancing cell-material interactions.
This study was supported by a grant of the Korea Healyhcare technology R&D project, ministry for health & welfare, Republic of korea (A120878).
- Branemark PI, Hansonn BO, Adell R, Breine U, Lindstrom J, Hallen O, et al. Osseointegrated titanium implants in the treatment of the edentulous jaw. Scand J Plast Reconstr Surg. 1977;11:171–5.Google Scholar
- Albrektsson T, Branemark PI, Hanson HA, Lindstrom J. Osseointegrated titanium implants. Requirements for ensuring a long-lasting, direct bone anchorage in man. Acta Orthop Scand. 1981;52:155–70.View ArticleGoogle Scholar
- Liu X, Chu PK, Ding C. Surface modification of titanium, titanium alloys, and related materials for biomedical applications. Mater Sci Eng R. 2004;47:49–121.View ArticleGoogle Scholar
- Molenberg A, Schwarz F, Herten M, Berner S, de Wild M, Wieland M. Improved osseointegration of a novel, hydrophilic Ti surface–a review. Mat-wiss u Werkstofftech. 2009;40:31–5.View ArticleGoogle Scholar
- Schliephake H, Scharnweber D. Chemical and biological functionalization of titanium for dental implants. J Mater Chem. 2008;18:2404–14.View ArticleGoogle Scholar
- Botticelli D, Berglundh T, Buser D, Lindhe J. The jumping distance revisited: an experimental study in the dog. Clin Oral Implant Res. 2003;14:35–42.View ArticleGoogle Scholar
- Rosales-Leala JI, Rodríguez-Valverdeb MA. Effect of roughness, wettability and morphology of engineered titanium surfaces on osteoblast-like cell adhesion. Colloids and Surfaces A: Physicochem Eng Aspects. 2010;365:222–9.View ArticleGoogle Scholar
- Han IH, Kwon BJ, Vagaska B, Kim BJ, Kang JK, Lee MH, et al. Nitrogen grafting onto polycarprolactone by a simple surface modification with atmospheric pressure glow discharge (Ar-APGD) and promoted neonatal human fibroblast growth. Macromol Res. 2011;19(11):1134–41.View ArticleGoogle Scholar
- Park BJ, Lee DH, Park JC. Sterilization using a microwave-induced argon plasma system at atmospheric pressure. Phys Plasmas. 2003;10:4539–44.View ArticleGoogle Scholar
- Park JC, Park BJ, Han DW, Lee DH, Lee IS, Hyun SO, et al. Fungal sterilization using microwave-induced argon plasma at atmospheric pressure. J Microbiol Biotechnol. 2004;14(1):188–92.Google Scholar
- Lee KY, Park BJ, Lee DH, Lee IS, Hyun SO, Chung KH, et al. Sterilization of Escherichia coli and MRSA using microwave-induced argon plasma at atmospheric pressure. Surf Coat Tech. 2005;193:35–8.View ArticleGoogle Scholar
- Park BJ, Takatori K, S-Konishi Y, Kim IH, Lee MH, Han DW, et al. Degradation of mycotoxins using microwave-induced argon plasma at atmospheric pressure. Surf Coat Tech. 2007;201:5733–7.View ArticleGoogle Scholar
- Ryu GH, Yang WS, Roh HW, Lee IS, Kim JK, Lee GH, et al. Plasma surface modification of poly (D, L-lactic-co-glycolic acid) (65/35) film for tissue engineering. Surf Coat Tech. 2005;193:60–4.View ArticleGoogle Scholar
- Lim HR, Baek HS, Lee MH, Woo YI, Han DW, Han MH, et al. Surface modification for enhancing behaviors of vascular endothelial cells onto polyurethane films by microwave-induced argon plasma. Surf Coat Tech. 2008;202:5768–72.View ArticleGoogle Scholar
- Baek HS, Park YH, Ki CS, Park JC, Rah DK. Enhanced chondrogenic responses of articular chondrocytes onto porous silk fibroin scaffolds treated with microwave-induced argon plasma. Surf Coat Tech. 2008;202:5794–7.View ArticleGoogle Scholar
- Bachmann J, Ellies A, Hartge KH. Development and application of a new sessile drop contact angle method to assess soil water repellency. J Hydrol. 2000;231–232:66–75.View ArticleGoogle Scholar
- Van Oss CJ, Giese Jr RF, Good RJ. Re-evaluation of the surface tension components and parameters of polyacetylene from contact angles of liquids. Langmuir. 1990;6:1711–3.View ArticleGoogle Scholar
- Vandrovcova M, Vacik J, Svorcik V, Slepicka P, Kasalkova N, Vorlicek V, et al. Fullerene C60 and hybrid C60/Ti films as substrates for adhesion and growth of bone cells. Phys Stat Sol (a). 2008;205(9):2252–61.View ArticleGoogle Scholar
- Han IH, Barbora V, Seo HJ, Kang JK, Kwon BJ, Lee MH, et al. Promoted cell and material interaction on atmospheric pressure plasma treated titanium. Appl Surf Sci. 2012;258:4718–23.View ArticleGoogle Scholar
- Anand V, Gowravaram MR. On the purity of atmospheric glow-discharge plasma. IEEE Trans Plasma Sci. 2009;37:1811–6.View ArticleGoogle Scholar
- Takahashi T, Ruan JZ, Kubota S, Shiraishi F. Electron thermalization in argon-nitrogen gas mixture excited by 252Cf fission fragments. Phys Rev A. 1982;25:2820–3.View ArticleGoogle Scholar
- Sun WT, Li G, Li HP, Bao CY, Wang HB, Zeng S, et al. Characteristics of atmospheric-pressure, radio-frequency glow discharges operated with argon added ethanol. J Appl Phys. 2007;101:123302.View ArticleGoogle Scholar
- Huang C, Yu QS, Hsieh F-H, Duan YX. Bacterial deactivation using a low temperature argon atmospheric plasma brush with oxygen addition. Plasma Process Polym. 2007;4:77–87.View ArticleGoogle Scholar
- Shen H, Xixue H, Yang F, Bei J, Wang S. Combining oxygen plasma treatment with anchorage of cationized gelatin for enhancing cell affinity of poly (lactide-co-glycolide). J Biomaterials. 2007;28(29):4219–30.View ArticleGoogle Scholar
- Davies JE. Understanding peri-implant endosseous healing. J Dent Educ. 2003;67(8):932–49.Google Scholar
- Han IH, Barbora V, Park BJ, Lee MH, Lee SJ, Park JC. Selective fibronectin adsorption against albumin and enhanced stem cell attachment on helium atmospheric pressure glow discharge treated titanium. J Appl Phys. 2011;109:124701–6.View ArticleGoogle Scholar
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.